Is the adapter for 384 well plates autoclavable?
Yes, the adapter for 384 well plates is autoclavable at 121 °C and 1 bar overpressure for 20 minutes.
Is it possible to dispens liquids in free-jet with the epMotion 96?
No, the epMotion 96 should not be used with free-jet dispensing. It is recommended to dispens directly into liquid.
How can I pipet 1 µL with the epMotion 96?
For transferring 1 µL with the epMotion 96 we recommend to use the "small volume" mode.
What can I do to prevent the O-rings of the epMotion 96 from erosion?
To protect the O-rings of the epMotion 96 from erosion we recommend to lubricate them with a thin layer of silicone spray.
How can I transfer volatile liquids (ethanol) with the epMotion 96 without drop formation on the tips?
We recommend to pre-wet the tips of the epMotion 96 via 2-3 aspration and dispensing cycles to avoid drop formation on the tips. Furthermore a low speed level (1-5) is advantageous. More information are available in White paper 39
Does a chemical resistance sheet for the epMotion 96 exist?
Yes you may download a chemical resistance sheet
from the homepage.
How can I transfer viscous liquids with the epMotion 96?
We recommend a very low aspiration and dispensing speed (1-3) when working with viscous liquids and the mode "Reverse pipetting". More information are available in White paper 39
How do I multidispense small volumes of 1-5 µL?
When dispensing small volumes of 1-5 µL some points have to be considered:
1) Use 50 µL tips
2) Always dispense into liquid or on a liquid surface
3) Use speed 9 for dispensing
4) Immerse the tip 2-3 mm into the liquid, press dispense and slowly remove the tip from the liquid
A pop-up window asks me to perform a firmware update after I staretd my iPod. How can I perform the firmware update?
Performing the firmware update is mandatory when the pop-up window showed, otherwise accurate pipetting and multiple functions on the epMotion96 and 96xl are not available. A firmware update is easily done by following the steps in short protocol 27
"How to perform a firmware update on epMotion 96 and epMotion 96xl".
Is it possible to use 10 µL epT.I.P.S. motion reloads with epMotion 96?
No, it is impossible to use the 10 µL epT.I.P.S. motion reloads with epMotion 96.
Is it possible to autoclave the Varispenser 2(x) attached to a bottle?
Yes, it is possible to autoclave the Varispenser 2(x) attached to a bottle (121 °C, 20 minutes, 1 bar overpressure).
Is it possible to exchange the dispensing tube of the Varispenser 2 and 2x among each other?
Yes, it is possible to exchange the dispensing tube of the Varispenser 2 with a dispensing tube of the Varispenser 2x. Please note that the device is then capable of recirculating the liquid back into the bottle.
Do aspiration tubes of different lengths exist for Varispenser 2(x)?
Yes, aspiration tubes of different lengths are available to cover most bottle types.
Does the aspiration tube of the Varispenser 2(x) need to be cut diagonal for aspiration?
No, the aspiration tube of Varispenser 2(x) does not need to be cut diagonal, because two holes on the bottom side of the tube enable liquid uptake.
What is the "flexible discharge tube" used for?
The "flexible discharge tube" can be used to reach into small or narrow-standing tubes without tilting the botlle. More Information are available in the White Paper 38
For which liquids a drying tube is beneficial?
The drying tube should be used when working with fuming acids to avoid dilution. It can tehn be filled with e.g., calcium chloride or silica pills. When working with alkaline solutions filling of the drying tube with sodium hydroxide pills prevents carbonate formation and crystallization of the liquid. More Information are available in the White Paper 38
Is it possible to dispense sterile medium with the Varispenser 2(x)?
Yes, it is possible to dispense sterile medium with the Varispenser 2(x). We recommend the usage of a sterile filter at the back of the device to keep the sterility of the medium. More Information are available in the White Paper 38
Is it possible to adjust the Varispenser 2(x)?
Yes, it is possible to adjust the Varispenser 2(x) for liquids with different properties than water. Please refer to the operating manual for further information.
How can the Varispenser 2(x) be decontaminated?
The outer parts of the Varispenser 2(x) can be wiped with lint-free tissue and 70 % ethanol. When sterilization of the inner parts is needed, the Varispenser 2(x) can be autoclaved).
Is the Varispenser 2/2x suited for dispensing 100% acetic acid (glacial acetic acid)?
Yes. 100% acetic acid can be dispensed with the Varispenser 2/2x. When using concentrated acids the usage of a drying tube is furthermore highly recommended. More information about the chemical resistance of the Varispenser 2/2x is available here
Can the Varispenser 2/2x be used to dispense a conc. sulfuric acid which contains potassium bichromate and silver sulfate?
Yes, you can use the Varispenser 2/2x for this application. Please notice, when using concentrated acids a drying tube should be connected to the Varispenser. More information about the chemical resistance of the Varispenser 2/2x is available here
Which physical limits does the Varispenser 2/2x has?
The device is intended for dispensing liquids with following properties: ambient temperature 15°C - 40°C , maximum density: 2.2 g/cm3, maximum vapor pressure: 500 mbar. Above 300 mbar slow aspiration is recommended to avoid boiling., kinematic viscosity: 500 mm2/s. More details are written in the operating manual
I bought a new Varispenser 2(x) and drops are visible inside the device. Additionally an intense smell was present when opening the cardbox. What is that?
Each Varispenser 2(x) is calibrated before packaging using distilled water. Afterwards each device is rinsed with an ethanol-glycerol solution to dry faster. The drops inside the device and the smell occur from this solution. The device should be completely dry after a few minutes outside the packaging.
What is the maximum viscosity limit for ViscoTips?
ViscoTips have been tested with liquds up to 14.000 mPa*s which equals clear liquid honey.
Why should I pre-wet ViscoTip?
We recommend pre-wetting of ViscoTip to avoid air bubbles. Furthermore, the pipetting results will be more precise and accurate after pre-wetting.
How can ViscoTip aspirate creamy liquids (skin cream, curd)?
Creamy liquids such as skin cream or curd flow very slowly and holes tend to form when a part of the liquid is removed. Therefore we recommend aspiration of creamy liquids in a circular movement to avoid air intake.
Furthermore, the liquid can be centrifuged prior to aspiration.
How can I decontaminate the Move It lower part?
With Xplorer plus, the lower parts are autoclavable. With Research plus the complete pipette is autoclavable.
The adjustment of cone spacing of the electronic model is not motorized. What is the reason?
When working with plates and multi-channel pipettes it is a prerequisite to have a rotatable lower part. Fixed lower parts make the work inconvenient and non-ergomic. For this reason Eppendorf has invented a lower part combining the factors "rotating" with "adjustable cone spacing". The fact, that the lower part is not fixed means that the motor for the cone space adjustment cannot be integrated in the pipette's upper part. Having a rotating lower part requires a motor in the lower part. This extra weight would have made the pipette "foot-heavy" and becoming non-ergonomic. For this reason Eppendorf decided for a manual cone spacing.
The Move It lower parts show indications for plates. I have a plate which does not fit into these formats. Can I nevertheless use the Move It pipettes?
First, please check with the option table, if you have the suitable Move It pipette availabe. Maybe you simply need a different model for your applications.
In case you have an atypical vessel format you can of course also use Move It if not exceeding the distance range. Use a ruler to check which Move It model is suitable for your atypic vessel type. Within this range all distances can be set. Therefore please proceed as follows:
1) Attach tips
2) Set the two format limiters to maximum width range
3) Use the rotating knob to adjust the space between the tips until fitting to the atypic vessel
4) Set one of the format limiter to this position
5) Set the rotating knob to the second vessel type
6) Set the format limiter to this tip distance
You can now work between these set limits by simply using the rotating knob.
Could you please send me an overview over the chemical resistance of the Move It lower parts?
The Move It lower parts are exclusively made of the high-tech materials we have already build in the
Research plus and Xplorer pipettes. Thus the chemical resistance table for the Research plus applies also for both, the Xplorer Move It lower parts and the Research plus Move It lower parts (but not for the upper part of Xplorer due to the electronics). The list is available on the eppendorf website (Research plus / Operating Manual)