이 사이트에서 나가려고 합니다.
현재 장바구니는 아직 저장되지 않았고 새로운 사이트에서 또는 다시 방문할 때 복원할 수 없습니다. 장바구니를 저장하려면 계정에 로그인하십시오.
How to Stop Dripping When Pipetting Volatile Liquids
Who isn’t aware of acetone, ethanol & co. starting to drip out of the pipette tip directly after aspiration? Probably, every one of us has experienced this. Supposed secret recipes like “working as fast as possible” while “placing the tubes very close to each other to avoid chemical loss and spillage” belong to your daily practices? Even if chemical droplets ran faster it is relatively often tolerated that pipetting is not accurate anymore. Just some small changes in pipetting techniques, and the right choice of pipette type can help overcome these daily challenges!
Why do pipettes drip?Classic pipettes start dripping when pipetting volatile liquids due to the air inside the pipette. This so-called air cushion exists between the sample liquid and the piston inside the pipette. As commonly known, air is flexible and adapts to external influences such as temperature and air pressure by expanding or compressing. Liquids are also subject to external influences and naturally evaporate as the humidity of air is lower. A volatile liquid evaporates much faster than water. During pipetting, it evaporates into the air cushion forcing the latter to expand and liquid is pressed out of the pipette tip … the pipette drips.
How to prevent liquids from dropping outOne technique to reduce or even stop the dripping is to achieve a higher percentage of humidity in the air cushion. This is done by pre-wetting the pipette tip and thereby saturating the air cushion. When using low volatile liquids such as 70 % Ethanol or 1 % acetone, aspirate and dispense the sample liquid a minimum of 3 times, before aspirating the sample volume you want to transfer. If the concentration of volatile liquid is higher, repeat these pre-wetting cycles 5-8 times. However, with very high concentrations such as 100 % ethanol or chloroform, this won’t be enough. It’s best to use another type of pipette: a positive displacement pipette. These pipettes use tips with an integrated piston without an air cushion. The sample is in direct contact with the piston and there is no risk of dripping.
Become a master of pipettingYou can easily improve your accuracy when pipetting volatile liquids by choosing the correct technique or changing the tool you are using. Additionally, you will increase safety by avoiding spillage and simplify your workflow. If you want to learn more on correct pipetting, watch our Youtube video tutorial How to pipette correctly .
- Legal Notice
- Trademarks & Patents
- Cookie Info
- For commercial customers only. All prices VAT exclusive.
- 상호: 에펜도르프코리아 주식회사 대표자: Christian Wolfgang Groeger 주소: 서울시 강남구 테헤란로 322 서관 6층 (역삼동, 한신인터밸리 24빌딩) 전화번호: 1577-4395 전자우편주소: email@example.com 사업자등록번호: 220-88-27824 통신판매업번호: 2023-서울강남-00526 판매 및 납품 일반약관
- © 2023 Eppendorf SE