What does Advanced Maintenance of an incubator include?
Advanced Maintenance includes checking the installation environment, the mains/power supply and gas supply and connections, replacing the in-line gas filter(s) and auto-zero filters and checking the door seals and door hinges. It also includes an inspection of the interior assemblies and a function test of the alarms. The operating parameters are checked against the manufacturer’s information and corrected, if required.
What does the Premium Service of an incubator include?
In addition to the comprehensive services of Advanced Maintenance, the Premium Service includes discounts on parts, labor and travel.
What is IQ or OQ and what do I need them for?
Installation and operational qualification; the IQ documents that the device has been delivered and installed correctly. The OQ checks the correct functioning of a device. IQ/OQ ensure that your quality management requirements are met and that your device operates properly and in accordance with the manufacturer’s specifications.
How often does the OQ need to be carried out for an incubator?
The company is responsible for technical checks of devices. Eppendorf recommends regular device qualifications depending on the intensity of use, however, they should be carried out at least once a year.
How can my incubator be cleaned/ disinfected?
Use a solution of 70% isopropanol (isopropyl alcohol) and 30% distilled water as a disinfectant. Follow the cleaning guidelines carefully that are described in your user manual.
Which parts of the incubator should be replaced at regular intervals?
Replace the HEPA filter of the CO2 supply line and the HEPA filter with automatic zero setting * at least once a year. (* Auto-zero filter is only used in R-models.)
What types of training or online training do you offer to enable me to improve my skills?
Please visit the Eppendorf webpage about the Eppendorf Training Center or contact your local Eppendorf Service for further information and a quotation. www.eppendorf.com/training
Which consumables should be replaced at regular intervals?
O-rings of FemtoJet 4 injection tube and grip head, consumption indicator depending on intensity of use
How can I see that the O-rings need to be replaced?
If leaks become noticeable on the grip head, they need to be replaced.